A measurement of the maximum diameter over the length of the delivery system, from the proximal end of the mounted stent to the distal tip.
With the stent pre-mounted or hand crimped on the delivery system in accordance with the instructions for use, and the sheath (if any) in place, measure the maximum diameter over the length from the proximal end of the mounted stent to the distal tip of the delivery system.
The measurement instrument is a non-contacting calibrated optical system.
Diameter of the length of the device.
ASTM F2081 – 06(2017)
Standard Guide for Characterization and Presentation of the Dimensional Attributes of Vascular Stents
Intravascular catheters — Sterile and single-use catheters — Part 5: Over-needle peripheral catheters
Cardiovascular implants — Endovascular devices — Part 2: Vascular stents
ViVitro Labs Advantages
Many companies look to academia to handle their testing needs due to the reduced cost. With ViVitro Laboratory Services you gain the following advantages:
Time to Market
We understand how critical your timelines are. Our expertise and experience ensure your schedule is met and our commitments are delivered on time.
Free Initial Consultations
All of our studies are scoped via no-cost, no obligation, collaborative consultation.
We happily provide Non-Disclosure Agreements for all services, including initial consultation.