The purpose of this test is to determine the deformation of the device in response to a localized compressive force perpendicularly applied to the longitudinal axis of the device, and to determine whether the stent recovers its original geometry after testing.
After measuring the stent outside diameter, the sample is positioned on a specific bench fixture. The sample stands horizontally on a V shaped support. A piston, controlled by a stepper motor, is advanced to the device until we obtain a preload that ensures point of contact at a pre-defined location on the specimen with stent. Then, the displacement is applied at a fixed rate while measuring the resultant force. The test stops when the displacement reaches defined target value. Then the piston is moved back to acquire elastic behavior data during sample unloading. At the end of the test, piston displacement and force are recorded.
A physiological temperature of 37°C can be used for thermo-sensitive material
Optical gauging machine
Automatic compression system
Balloon expandable stents
Self expanding stents
Cardiovascular implants — Endovascular devices — Part 2: Vascular stents
ViVitro Labs Advantages
Many companies look to academia to handle their testing needs due to the reduced cost. With ViVitro Laboratory Services you gain the following advantages:
Time to Market
We understand how critical your timelines are. Our expertise and experience ensure your schedule is met and our commitments are delivered on time.
Free Initial Consultations
All of our studies are scoped via no-cost, no obligation, collaborative consultation.
We happily provide Non-Disclosure Agreements for all services, including initial consultation.