Balloon Cyclic Fatigue
Balloons must withstand multiple inflations during clinical use to avoid inducing device failure or vessel damage.
Tested balloons that are not constrained by any test fixture or device are inflated in incremental steps to defined target pressure. Pressure is held for a specific interval of time. Balloons are repeatedly deflated and inflated to the target pressure for a total of multiple cycles. Any loss of pressure, due to failure of the balloon, shaft, or proximal or distal seals is reported as a test failure.
The test is performed with an automatic inflation/deflation system.
- Angioplasty balloon catheters
- PTA, PTCA balloon catheters
- Any interventional systems with balloon technology
- Any delivery system with balloon technology
Intravascular catheters — Sterile and single-use catheters — Part 1: General requirements
Cardiovascular implants — Endovascular devices — Part 2: Vascular stents
FDA Guidance: Non-Clinical Engineering Tests and Recommended Labeling for Intravascular Stents and Associated Delivery Systems – Guidance for Industry and FDA Staff
- Balloon Inflation Deflation Time
- Balloon Rated Burst Pressure (RBP)
- Burst Strength
- Burst Strength – Balloon Method
- Air Leakage During Aspiration – Catheter Hub Assembly
- Burst Pressure Under Static Conditions
- Torsional Bond Strength
- Corrosion – Immersion Test
ViVitro Labs Advantages
Many companies look to academia to handle their testing needs due to the reduced cost. With ViVitro Laboratory Services you gain the following advantages:
Time to Market
We understand how critical your timelines are. Our expertise and experience ensure your schedule is met and our commitments are delivered on time.
Free Initial Consultations
All of our studies are scoped via no-cost, no obligation, collaborative consultation.
We happily provide Non-Disclosure Agreements for all services, including initial consultation.